|Product||Name: Artemis V6 UVC Disinfection Robot
Weight: 125.6 lbs
Size: 17.7" x 17.7" x 57.55"
|Lamp||Model: Signify (PHILIPS TUV T8)
power: 30W * 6 =180W
Length: 3 ft
||Type: 10S16P 20,800 mAh/37V
Last: 2.5~3 hours
OUTPUT: DC 42V 5A
| Charging Method
DC Direct Charge
||Output Voltage: 42V DC
Output Current: 5A
Artemis V6 Disinfection Robot
Automatic Disinfection Cycle
Set a disinfection schedule for automatic disinfection route completion
Autonomous planning of disinfection routes. Disinfect an entire area or a specific target point
When the battery life is at the minimum, the Artemis V6 will automatically return to the charging station
6 Bulb PHILIPS UVC lamp, High intensity UVC disinfection
Illumination power: 180W 254nm ultraviolet shortwave Virus inactivation rate: 99.9999%
Local area network remote control via web browser
Hospitals & Other Medical Facilities
Public Transportation Hubs
Operating system Android（5.1） Processor type RK3128 Built-in clock RTC RAM LPDDDR3 1GB Built-in NAND FLASH 8GB External I/O port USB 2.0 (MICROUSB) Emergency stop button Yes Power button One-Key Boot Screen 7 inches, IPS screen (16:9), 1024 x 600 Language support English, Chinese Speaker 4Ω/3W Mono Speaker Network Dual band 2.4GHz & 5GHz WIFI 802.11b/g/n/ac
Operating system LINUX RAM 4GB Motor driven 5.5 inch hub motor Gyro 9 - Axis High-Precision Attitude Sensor LiDAR Max Range: 65.6 ft (20 m)
Wave Length: 905 nm
Horizontal FOV: 230°
Network Dual Band 2.4GHz & 5GHz WIFI 802.11b/g/n/ac Processor type ITX-H45-I526L VER: 1.1A(I5-4300U) Hard disk 32GB High-Speed Solid State Drive
When using UVC sterilizating robots, people and animals should stay away from the ultraviolet radiation area. Do not look directly at the ultraviolet light source.
The best disinfection temperature is 5°F - 104°F, the humidity is less than 80%. If the temperature is too high or too low, sterilization will be effected.
During the use of the UV lamp, the radiation intensity gradually decreases. The UV radiation ﬂux should be measured regularly, and it is recommended to replace it after 8000 hours.
When using the ultraviolet disinfection robot for disinfection, keep the lamp tube clean, and clean the surface of the lamp tube when the power is oﬀ.
After using the ultraviolet disinfection robot to disinfect, it is recommended to ventilate the room for more than 30 minutes.